To compare tendon-bone interface motion and cyclic loading in a single-row, triple-loaded anchor repair with a suture-tape, rip-stop, double-row rotator cuff repair.
Using 18 human shoulders from 9 matched cadaveric pairs, we created 2 groups of rotator cuff repairs. Group 1 was a double-row, rip-stop, suture-tape construct. Group 2 was a single-row, triple-loaded construct. Before mechanical testing, the supraspinatus footprint was measured with calipers. A superiorly positioned digital camera optically measured the tendon footprint motion during 60° of humeral internal and external rotation. Specimens were secured at a fixed angle not exceeding 45° in reference to the load. After preloading, each sample was cycled between 10 N and 100 N for 200 cycles at 1 Hz, followed by destructive testing at 33 mm/s. A digital camera with tracking software measured the repair displacement at 100 and 200 cycles. Ultimate load and failure mode for each sample were recorded.
The exposed anterior footprint border (6.5% ± 6%) and posterior footprint border (0.9% ± 1.7%) in group 1 were statistically less than the exposed anterior footprint border (30.3% ± 17%) and posterior footprint border (29.8% ± 14%) in group 2 (P = .003 and P < .001, respectively). The maximal internal rotation and external rotation tendon footprint displacements in group 1 (1.6 mm and 1.4 mm, respectively) were less than those in group 2 (both 3.6 mm) (P = .007 and P = .004, respectively). Mean displacement after 100 cycles for group 1 and group 2 was 2.0 mm and 3.2 mm, respectively, and at 200 cycles, mean displacement was 2.5 mm and 4.2 mm, respectively (P = .02). The mean ultimate failure load in group 1 (586 N) was greater than that in group 2 (393 N) (P = .02). The suture-tendon interface was the site of most construct failures.
The suture-tape, rip-stop, double-row rotator cuff repair had greater footprint coverage, less rotational footprint displacement, and a greater mean ultimate failure load than the triple-loaded, single-row repair on mechanical testing. No double-row or single-row constructs showed 5 mm of displacement after the first 100 cycles. The most common failure mode for both constructs was suture tearing through the tendon.
Differences in cuff fixation influence rotational tendon movement and may influence postoperative healing. Stronger repair constructs still fail at the suture-tendon interface.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Arthroscopy
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Arthroscopic single-row supraspinatus tendon repair with a modified mattress locking stitch: A prospective, randomized controlled comparison with a simple stitch.Arthroscopy. 2008; 24: 1005-1012
- The time for functional recovery after arthroscopic rotator cuff repair: Correlation with tendon healing controlled by computed tomography arthrography.Arthroscopy. 2008; 24: 25-33
- The biomechanical effects of dynamic external rotation on rotator cuff repair compared to testing with the humerus fixed.Am J Sports Med. 2007; 35: 1931-1939
- Biomechanical comparison of a single-row versus double-row suture anchor technique for rotator cuff repair.Am J Sports Med. 2006; 34: 407-414
- Arthroscopic single-row versus double-row suture anchor rotator cuff repair.Am J Sports Med. 2005; 33: 1861-1868
- Repairs of the rotator cuff.J Bone Joint Surg Am. 1991; 73: 982-989
- Clinical outcome after structural failure of rotator cuff repairs.J Bone Joint Surg Am. 2000; 82: 304-314
- Repair site integrity after arthroscopic transosseous-equivalent suture-bridge rotator cuff repair.Am J Sports Med. 2008; 36: 1496-1503
- A new approach to improving the tissue grip of the medial-row repair in the suture-bridge technique: The “modified lasso-loop stitch”.Arthroscopy. 2009; 25: 691-695
- The effect of the angle of suture anchor insertion on fixation failure at the tendon-suture interface after rotator cuff repair: Deadman's angle revisited.Arthroscopy. 2009; 25: 597-602
- A novel method to determine suture anchor loading after rotator cuff repair—A study of two double-row techniques.Bull NYU Hosp Jt Dis. 2010; 68: 25-28
- Mechanical strength of arthroscopic rotator cuff repair techniques: An in vitro study.J Bone Joint Surg Am. 2002; 84: 2152-2160
- Knotless rotator cuff repair in an external rotation model: The importance of medial-row horizontal mattress sutures.Arthroscopy. 2011; 27: 471-478
- Biomechanical advantages of triple-loaded suture anchors compared with double-row rotator cuff repairs.Arthroscopy. 2010; 26: 316-323
- Comparative testing by cyclic loading of rotator cuff suture anchors containing multiple high-strength sutures.Arthroscopy. 2010; 26: S134-S141
- Evaluation of 5 knots and 2 suture materials for arthroscopic rotator cuff repair: Very strong sutures can still slip.Arthroscopy. 2006; 22: 38-43
- Cyclic load and failure behavior of arthroscopic knots and high strength sutures.Arthroscopy. 2009; 25: 192-199
- Part I: Footprint contact characteristics for a transosseous-equivalent rotator cuff repair technique compared with a double-row repair technique.J Shoulder Elbow Surg. 2007; 16: 461-468
- Part II: Biomechanical assessment for a footprint-restoring transosseous-equivalent rotator cuff repair technique compared with a double-row repair technique.J Shoulder Elbow Surg. 2007; 16: 469-476
- Tendon-bone interface motion in transosseous suture and suture anchor rotator cuff repair techniques.Am J Sports Med. 2005; 33: 1667-1671
- Suture anchor insertion angle and the deadman theory.Arthroscopy. 2009; 25 (author reply 1365-1366 (letter)): 1365
- Does slower rehabilitation after arthroscopic rotator cuff repair lead to long-term stiffness?.J Shoulder Elbow Surg. 2010; 19: 1034-1039
- Rehabilitation of the rotator cuff: An evaluation-based approach.J Am Acad Orthop Surg. 2006; 14: 599-609
- Exercise in the treatment of rotator cuff impingement: A systematic review and a synthesized evidence-based rehabilitation protocol.J Shoulder Elbow Surg. 2009; 18: 138-160
- Arthroscopic rotator cuff repair: Scientific rationale, surgical technique, and early clinical and functional results of a knotless self-reinforcing double-row rotator cuff repair system.J Shoulder Elbow Surg. 2010; 19: 83-90
- Arthroscopic supraspinatus tendon repair with suture-bridging technique: Functional outcome and magnetic resonance imaging.Am J Sports Med. 2010; 38: 983-991
- The effect of abduction and rotation on footprint contact for single-row, double-row, and modified double-row rotator cuff repair techniques.Am J Sports Med. 2009; 37: 1599-1608
- Cyclic loading of anchor-based rotator cuff repairs: Confirmation of the tension overload phenomenon and comparison of suture anchor fixation with transosseous fixation.Arthroscopy. 1997; 13: 720-724
Published online: May 17, 2012
Accepted: February 6, 2012
Received: July 25, 2011
The authors report the following potential conflict of interest or source of funding in relation to this article: This study was supported by a grant from Arthrex, Naples, FL. The anchors were provided and used.
© 2012 Arthroscopy Association of North America. Published by Elsevier Inc. All rights reserved.